pyafm: tools for controlling atomic force microscopes.
Pyafm is a set of tools for controlling atomic force microscopes. It provides control of AFM postition using both short-range (piezo) and long range (stepper) vertical positioning. There are separate modules for controlling the piezo (pypiezo) and stepper (stepper), this module mostly contains methods that require the capabilities of both.
This module can optionally include temperature sensing via a pypid backend.
# emerge -av app-portage/layman # layman --add wtk # emerge -av sci-physics/pyafm
Pyafm requires the following Python modules:
Getting the source
Pyafm is available as a Git repository:
$ git clone git://tremily.us/pyafm.git
There are also periodic bundled releases. For example, get version 0.3 as a gzipped tarball with:
$ wget 'http://git.tremily.us/?p=pyafm.git;a=snapshot;h=v0.3;sf=tgz' $ tar -xzvf pyafm-0.1.tar.gz
After downloading, change to the source directory and run:
$ python setup.py install
to install pyafm. Run:
$ python setup.py install --help
to see a list of installation options you may want to configure.
The docstrings include some pretty detailed tests to get you started. dir() and help() are your friends ;). One neat feature that I’ve added recently (2012-03-16) is the ability to store and load complete AFM configurations via h5config. For example, when you first use pyafm, it may take you a bit to dig up all the necessary calibration constants, etc. and plug them into your config:
>>> import pycomedi.constant >>> import pypiezo.config >>> import pyafm.config >>> config = pyafm.config.AFMConfig() >>> config['name'] = '1B3D9' >>> config['main-axis'] = 'z' >>> config['piezo'] = pypiezo.config.PiezoConfig() >>> config['piezo']['name'] = '2253E' >>> config['piezo']['axes'] = [ ... pypiezo.config.AxisConfig(), pypiezo.config.AxisConfig()] >>> config['piezo']['axes']['gain'] = 20 >>> config['piezo']['axes']['sensitivity'] = 8.8e-9 >>> config['piezo']['axes']['channel'] = pypiezo.config.OutputChannelConfig() >>> config['piezo']['axes']['channel']['analog-reference] = pycomedi.constant.AREF.ground >>> config['piezo']['axes']['channel']['analog-reference] = pycomedi.constant.AREF.ground
and on, and on ;). Don’t worry though, once you finish telling Python about your particular AFM configuration, you can get a working AFM instance quite easily:
>>> devices =  >>> afm = AFM(config=config, devices=devices)
That takes care of opening all the channels and initializing all the pieces you configured above. devices will end up full of any pycomedi.device.Device instances that you need. Once you’re happy with your setup, make sure the config object is up-to-date with:
Which copies any new object state into the .config attribute (e.g. bit-to-volt conversion polynomials). Then save your configuration with:
>>> import pyafm.storage >>> pyafm.storage.save_afm(afm=afm, filename='whatever', group='/optional')
Which will create a HDF5 file at the specified path, and store the AFM configutaion under the specified group. Both filename and group are optional. If you leave them out, they will default to ~/.config/pyafm-default.h5 and / respectively.
The next time you need to do something with the AFM, just load your old config file. If you used the default location, that’s as easy as:
>>> import pyafm.storage >>> devices =  >>> afm = pyafm.storage.load_afm(devices=devices) >>> afm.load_from_config()
When you’re done using them, it’s good practice to close any devices in devices:
>>> for device in devices: ... device.close()
Using load_afm with the default path is a good way to keep your AFM configuration synchronized across several applications. That way, there’s only one place you need to update if you recalibrate your piezo or rebuild an amplifier.
Download the file for your platform. If you're not sure which to choose, learn more about installing packages.